ONICON’s F-4300 Clamp-On Ultrasonic Flow Meters have various transducer options available. ONICON provides a dielectric coupling compound with every meter shipped. Applying coupling compound to the transducers can improve the overall water resistance and signal quality.
Application to New Transducers
Prior to installing a new F-4300 and the transducers, apply a small amount of the provided dielectric coupling compound to the inside of the transducer. Refer to the image below.
Application to Existing Transducers
Turn off the power that connects to the transducers. Remove the transducers and apply a small amount of the provided dielectric coupling compound to the inside of the transducer. Refer to the image below.
If you should have any technical questions about your ONICON flow meter installation, please contact ONICON Technical Support at (727) 447-6140 or email@example.com.Download This Document